Over the past decade, options in prosthetic feet have expanded to
more than 50 different models. Today, amputees have a wide array of
feet to choose from, designed for walking, dancing, cycling, golfing,
swimming, snow skiing and running. Heavier wood and steel materials
have been replaced by lightweight plastics, aerospace alloys and
carbon-fiber composites. Much like the human foot , many of today’s
prosthetic feet can store and return some of the energy generated
during walking. Other key attributes include toe and heel springs
that allow more natural movement at the ankle, shock absorption,
multiaxial rotation, adjustable heel heights and waterproof
How can you select the right foot for your
lifestyle? A number of factors must be
considered, including your amputation level, age,
weight, foot size, activity level and occupational
Structurally, prosthetic feet can be divided into
two groups: those with a rigid connection to the
prosthetic shank (nonarticulated) and those with
a hinged ankle mechanism (articulated)09/10/2006be categorized in
the following groups:
- Elastic (flexible) keel
SACH and Elastic Keel Feet. The most basic
prosthetic feet come in two types: Solid
Ankle Cushioned Heel (SACH) and elastic keel
configurations. These designs consist of crepe
neoprene or urethane foam molded over an
inner keel and shaped to resemble a human
foot. Because they have no hinged parts, these
basic feet are inexpensive, durable and virtually
maintenance-free. These feet offer cushioning
and energy absorption but do not store and
return the same amount of energy as dynamicresponse
feet do. SACH and elastic keel feet are
designed for amputees who do a limited amount
of walking with little variation in speed. They
are used most often on a first or preparatory
prosthesis and are often later replaced by a
more dynamic type of foot.
The SACH is the simplest type of
nonarticulated foot. The name refers to a rubber
heel wedge that provides a minimal amount of
ankle action by compressing under load during
the early part of the stance phase of walking.
The keel is rigid, which provides midstance
stability but little lateral movement. The SACH
is available in various heel heights so it may be
worn with different shoes.
Elastic (flexible) keel feet allow motion
similar to that of SACH feet. In addition,
they allow the forefoot to conform to
uneven terrain but still remain stiff and
stable during standing and walking.
Articulated prosthetic feet may be
09/10/2006xis” refers to motion in one or more
of three different planes, similar to the
movement of the natural foot:
Internal and external rotation — The
clockwise and counterclockwise
rotations of the leg while the foot is in
contact with the ground.
Inversion and eversion — The inward
and outward (side to side) motions of
the ankle and leg while walking.
Dorsiflexion and plantarflexion — The
up and down movements at the ankle
that enable the leg to move forward
over the foot, pushing the forefoot to
Prosthetic feet that have movement in
two or three axes provide increased
mobility at the ankle, which helps
accommodate uneven surfaces.
||Single-Axis Feet. The articulated single-axis foot contains an ankle joint that allows the foot to move up and down, enhancing knee stability. The more quickly the full sole of the foot is in contact with the ground, the more stable the prosthesis becomes. This is beneficial for users with higher levels of amputation, from the knee to the hip. The wearer must actively control the prosthesis to prevent the knee from buckling, and the single-axis ankle-foot mechanism reduces the effort required. Unfortunately, the single-axis ankle adds weight to the prosthesis, requires periodic servicing and is slightly more expensive than the most basic SACH foot. For these reasons, it is most appropriate for individuals whose major concern is stability.
||Multiaxis Feet. Although similar to the single-axis in terms of weight, durability and cost, the multiaxis foot conforms better to uneven surfaces. In addition to the up and down mobility of the singleaxis, the multiaxis foot can also move from side to side. Since the added ankle motion absorbs some of the stresses of walking, this helps protect both the skin and the prosthesis from wear and tear. It has been particularly well-received by hikers, golfers, dancers and others who need significant mobility for their work or recreational needs.
Dynamic-Response Feet. People with more active lifestyles typically prefer these more responsive feet. They are ideal for amputees who can vary their walking speed, change direction quickly or walk long distances. Dynamicresponse feet store and release energy during the walking cycle. They provide a subjective sense of push-off for the wearer, a more normal range of motion and a more symmetric gait. Some dynamic-response feet feature a split-toe design that further increases stability by mimicking the inversion/ eversion axis of the human ankle and foot. The comfort and responsiveness of a dynamic foot can also encourage an individual to advance from a minimal activity level to a moderate activity level. Further, some have been shown to reduce impact on the heel of a sound foot.
Just as there is no single tool that is perfect for every job, there is no single foot that is perfect for every amputee. Knowing your options will enable you to discuss this issue clearly with your prosthetist. Evaluate the pros and cons of different feet so you can make the best choice for your individual aspirations and abilities.
Disclaimer: The views and opinions expressed in this publication are those of the authors and are not necessarily those of the Amputee Coalition, the Department of the Army, the Army Medical Department, or any other agency of the US Government.
|Back to Top